Great for kids, hobbyists, and professionals, the Plugable USB2-MICRO-250X Digital Microscope allows you to work quickly and effectively at the microscopic level. Common uses include classroom exploration, stamp or coin analysis, micro-soldering, garden parasite identification, and much more. The companion software available on our website for Windows and OS X allows you to control aspects of the camera functionality directly from your computer.
Capture resolution and exposure can be changed with a few clicks, and video or time-lapse modes allow for continuous observation of a subject. Interface USB 2.0 Magnification Range 0x-250x Illumination 4 SMD LEDs (dimmable) Sensor Type CMOS Maximum Resolution 1600 x 1200 snapshot 640 x 480 video Controller EtronTech eSP570A Frequently Asked Questions Q: How do I magnify an object? A: The level of magnification is controlled by a combination of the distance to the object and the focus point set by the grey focus wheel. The closer the object, the higher the magnification possible. Q: I can’t get my microscope to zoom past a very low magnification, how do i get to 250x?
A: At the closest distance with the clear shield touching the object, there are two possible magnifications. The first is at around 50x, but if you continue to rotate the focus wheel, the image will get blurry, then sharp again at a much higher magnification.
Q: How do I dim the LED lights? A: The LED lights can be dimmed or turned off by rotating the grey LED light control wheel near the top of the microscope body. If you are having problems with glare, try viewing your intended object at an angle.
Q: When using the microscope, all I see is a black screen with no image. How can I fix this? A: From the factory, the LED control is in the off position. For most indoor lighting conditions, you will need to turn the LEDs on by rotating the grey light control wheel in order to see anything. Q: When I plug in the microscope, I see no image and the LEDs do not turn on.
Find software for your AmScope camera. We have a large selection of microscope cameras and digital microscopes, and this is where you find the software. We have imaging applications for Windows, Mac, and Linux.
What’s wrong? A: You must launch Digital Viewer or another webcam suite in order to use the microscope, nothing happens automatically on plug-in. Cara instal microsoft word.
The microscope’s LEDs won’t work unless the image sensor is actually being used. Q: I installed Digital Viewer on Windows, but I can’t figure out how to launch the software. What do I do? A: The installer creates a shortcut in your start menu, the default location being Start>Programs>Plugable Technologies (varies slightly depending on OS version). Many expect the installer to create a shortcut on the desktop, but we made the decision not to do this as we don’t want to clutter our customers desktop space without permission. Raam leela full movie youtube.
Q: When I open Digital Viewer, why do I only see output from my other webcam? A: The microscope must be selected in the device menu under settings.
The settings menu can be opened by clicking on the grey gear icon in the upper left of the window. Q: After a Windows Update, why does Digital Viewer tell me that my camera is already in use? A: There have been changes to how specific classes of devices are handled in Windows Privacy settings. We have a blog post detailing how to restore functionality here: Q: Why can’t I get this to work with my Android device? A: Android support is on a case by case basis.
Usb Microscope Software Windows 10
Intel pentium graphic drivers download. The manufacturer of the Android device must allow USB webcams or this will not work. Q: Will this work on my iPad/iPhone or other iOS device? A: No, this will not work. IOS devices do not have support for external USB cameras. Q: When I plug in the microscope, it is only listed as an Audio device and I have an Oculus Rift The microscope uses the same chipset as the Oculus DK2 sensors making the system believe it is just an additional Oculus sensor. The steps below should correct this problem!